I was wondering how to remove the extra gate on the C2- peak etc...
The NGC8206 will report 7 peaks for the heavy portion of the analysis (Chrom-1) by using calibration blends without neo-pentane. Components are the following: C6+, unmeasured lights, C3, IC4, NC4, IC5 and NC5. The neo-pentane (neoC5), if present, shall be in front of the IC5. The light portion of the analysis (Chrom-2) will show C3+, N2, CH4, CO2 and C2. I'll email a technical bulletin with the procedures on how to calibrate the analyzer without neo-pentane and how to remove the gate of the C2-.
My responsibility is to provide Technical Sales Support on Process Gas Chromatographs, Multiwave Photometers, RVP analyzers and Data Systems in ABB’s Measurement Products Business Unit and the markets it serves including refinery, petrochemical and chemical. I have been with ABB since January 2009 and held positions as Gas Chromatography and Process Photometers Specialist and Local Product Manager for ABB Italy since 2011. I’ve been appointed MED Region Technical Sales Support in 2011 and starting from 2014, I assumed a full time position as European Technical Sales Support for ABB Lewisburg Process Analytics. I hold a Bachelor’s and Master’s degree in Chemistry and a Ph.D. in Pure and Applied Chemistry from Università degli Studi del Piemonte Orientale “Amedeo Avogadro”.
If you want to purchase a product, need any pricing information or need help or support from ABB, please contact your local partner.